to create and rate content, and to follow, bookmark, and share content with other members.
N9923A Fieldfox (A.07.01) with Data Link 4.53
Question asked by
on Oct 28, 2013
on Oct 30, 2013 by am95405
Show 0 Likes
Hi, currently each marker has 3 decimal places, I will like to ask whether is it possible to reduce the marker value to just 1 decimal place instead ?
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Oct 30, 2013 10:11 AM
Can you be more specific about which part of the marker? I have attached a screenshot. The frequency part, or the amplitude?
Also, I assume you are referring to the Data link markers, not the markers on the analyzer itself.
Finally, what mode, the basic network analyzer mode?
The specifics would help us in defining the feature request.
Show 0 Likes
Retrieving data ...
GPIB/SCPI Commands wtih Agilent E8361A
SystemVue Connected Solution (Connected to Keysight Instruments)
Connecting SystemVue to Test Instruments for DPD Measurement
How to choose the right optical power meter for 850 nm signals
On a E4438C when using a PRAM file for data I was able to add a data bit with non burst too turn off the carrier. I can not seem to do this with the N5182B using the same PRAM file. How do I turn off the RF?