to create and rate content, and to follow, bookmark, and share content with other members.
54610B Ch1 waveform problem
Question asked by
on Jul 12, 2014
Show 0 Likes
Ch1 has low sensitivity and waveform display is noisy. It will display a waveform with a higher amplitude signal. Ch2 works great. Does anyone have schematics for this scope or any information that will help me diagnose this problem?
No one else has this question
Mark as assumed answered
This content has been marked as final.
(Required, will not be published)
Retrieving data ...
E4406A VSA software support
Spectrum Analyzer Models
Tim’s Blackboard #10: What Makes DFE a Nonlinear Equalizer?
Tim’s Blackboard #9: Eye-opening Experience with FFE
Firmware upgrade for Agilent 54825A oscilloscope