Signal Generators and Sources
to create and rate content, and to follow, bookmark, and share content with other members.
Issues with TME and E8257D
Question asked by
on Mar 25, 2014
on Apr 1, 2014 by tabbott
Show 0 Likes
I have a 50GHz E8257D that I am trying to self-maintain, and is throwing a System.OutOfMemoryException error on certain tests. The main test I need to complete is the High Band Power Flatness calibration and this pops up every time. Any ideas?
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Apr 1, 2014 8:39 AM
The “'System.OutOfMemoryException' was thrown.” is a known error. It is caused by having Microsoft .NET 4.5 Framework loaded onto the PC with TME. TME N7800A E.03.50 is not compatible with this version of .NET 4.5 Framework. We are currently working on the fix which estimated to be released by the end of July 2014.
To correct the issue:
1. Close TME.
2. Go the Control Panel’s “Programs and Features” section and uninstall Microsoft .NET 4.5 Framework.
3. Reboot the PC.
4. Now go to wwww.microsoft.com and download and install .Net 4.0 Framework. (both the Client Profile and Extended versions)
5. Please try running TME again.
Note: In rare cases an un-install and re-install on TME is required.
I hope this helped -
Show 0 Likes
Retrieving data ...
vrf User Changeable parameters
How to Control E3631A through C/C++
Executing and sending commands to console program
Power calibration with use of receiver "A"
Importing EMPro circuit component definition from ADS