to create and rate content, and to follow, bookmark, and share content with other members.
Current Measurement Resolution
Question asked by
on Oct 8, 2013
on Oct 9, 2013 by jdwindrvr
Show 0 Likes
I was wondering what the lowest current measurement the 34460A can make is. I see that there is a 100uA range, but no resolution or minimum signal specified (that I can find). Thank you.
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Oct 9, 2013 1:44 PM
The resolution of the 34460A/34461A is specified in terms of displayed digits (6 1/2), rather than absolute numbers. With 6 1/2 digits and the maximum aperture time of 100 plc, the displayed resolution for each current range is as follows:
100 uA 100 pA
1 mA 1 nA
10 mA 10 nA
100 mA 100 nA
1A 1 uA
3A 1 uA
Additional digits are returned beyond the resolutions listed if the DMM is programmed remotely.
1 person found this helpful
Show 0 Likes
Retrieving data ...
GND planes and GND references and Internal Ports
Distance between the layout pins for port 1 is electrically large
metal/via option in interdigitated capacitor sim
AEL in ADS 2012.08 (New User to AEL)
S-parameter port is electrically large