to create and rate content, and to follow, bookmark, and share content with other members.
Phase control on 35222A/B
Question asked by
on Apr 25, 2013
on May 10, 2013 by lhornburg
Show 0 Likes
Can someone confirm that the phase control function of the 33522 generators has a resolution of only 100mdeg and not 1mdeg as in the 33220/33250 units
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
May 10, 2013 1:45 PM
We just released a new firmware rev 2.09, that let's you set phase in degrees, radians, or time. Simply change the default units from the Units menu (front panel) or with the UNIT:ANGLe command. You can also select the unit when setting the phase via the front panel or remote interface.
If setting the phase in terms of time, you get far more resolution than you did previously with degrees. It also now lets you set the phase in terms of degrees down to 1mdeg.
The firmware can be downloaded here:
Show 0 Likes
Retrieving data ...
Tim's Back to Basics Seminar Workspace
Tim's HSD Seminar Workspace
Diodes 101 – What is a Diode?
Does My Power Supply Have Multiple Range Output?
How to trigger FieldFox when average on with SCPI commands