Data Acquisition and Switch Units
to create and rate content, and to follow, bookmark, and share content with other members.
What is the fastest scan rate of BLDL 3?
Question asked by
on Apr 25, 2013
on Apr 29, 2013 by tomc
Show 0 Likes
We have tried to scan at 100 samples per second and the software freezes.
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Apr 29, 2013 11:11 AM
What instrument and multiplexer card are you using? The one whose fastest scanning speed is spec'd at 100 rdgs/sec is the 34921A multiplexer for the 34980A. That speed is within one 20 channel bank. With BLDL3, the fastest I could get is 94 rdgs/sec. This is with the interval set to 0. I also tried it with the interval set to 10 ms. It did not freeze. I was connected with LAN sockets, which is the fastest IO connection. I'm running Windows Enterprize and Agilent IO Libraries 16.3.
Show 0 Likes
Retrieving data ...
ADS Example Book: Harmonic Balance (HB) Simulation
PNA-X B model Recall print colors command not working
Momentum and Schematic and RMSA
circuit component pin number does not match EM layout pin number
Advanced technical specifications for U1273A OLED Super yellow