to create and rate content, and to follow, bookmark, and share content with other members.
How to measure the channel power with E8362B?
Question asked by
on Jul 21, 2006
on Jul 24, 2006 by tsilchia001
Show 0 Likes
Hi,I am using E8257D and E8362B to measure the antenna pattern.The E8257D generates a RF signal as a transmitter,and the E8362B measures the channel power as a receiver.I want to know how to use the E8362B to measure the power?
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Jul 24, 2006 9:14 AM
It sounds as though your goal is a receiver calibration. In order to do this you must first perform a source calibration. The source calibration requires the use of a power meter to accurately characterize the actual stimulus power level from the analyzer. This corrected power is then used to build a correction table for the receiver during a receiver calibration. The E8362B analyzer will take control of the power meter via either the embedded GPIB or a USB to GPIB adapter, Agilent model number 82357A. The entire process is outlined in the PNA Help file. Hope this offers the assistance you were after.
Show 0 Likes
Retrieving data ...
Problem with allegro PCB extraction in ADS
Importing Cadence .brd files into ADS for evaluation?
BRD Import Error: The Cadence Allegro "extracta" utility was not found. Cadence Allegro PCB Designer must be installed to import a *.brd file.
Programming N6705B DC Power Analyzer
N6705B Data logger export to csv file