to create and rate content, and to follow, bookmark, and share content with other members.
E4446A Phase Noise Log Plot Measurement
Question asked by
on Oct 2, 2012
Show 0 Likes
I'm trying to make a log plot measurement while the unit is being vibe shocked. The LO doesn't unlock, but does jitter. I think the tracking algorithm can't keep up with the jitter. Any suggestions? Thanks.
No one else has this question
Mark as assumed answered
This content has been marked as final.
(Required, will not be published)
Retrieving data ...
Internal port calibration for SMD components
momentum SMD port calibration
Ports in Momentum: SMD vs TML
Custom time domain signal: from .csv file to .tim format
Introducing SystemVue 2016