to create and rate content, and to follow, bookmark, and share content with other members.
Impedance Matching 2 Ohms source to Antenna with 50 ohms VNA
Question asked by
on Jan 6, 2013
on Jan 10, 2013 by firstname.lastname@example.org
Show 0 Likes
VNA/Sim ISSUES --RESOLVED.
Edited by: rjrodrig on Jan 10, 2013 10:56 AM
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Jan 10, 2013 9:56 AM
Having forums like these are pointless and defeats the original purpose of having them in the first place. There are bunch of lurkers and no one cares to make a comment about a valid questions independently of how inane the question may be.
I found that the VNA measured impedance in a 50 ohms system is correct. There are no correction factors, just measurement errors. This was causing confusion in my simulations. Nothing to disregard as the original message was totally disregarded.
Edited by: rjrodrig on Jan 10, 2013 10:53 AM
Show 0 Likes
Retrieving data ...
Simulation problem in ADS software
Need help on Symbolically-Defined Devices
Default mesh settings
AEL syntax highlighting in Notepad++
How to import different Workspace and Data Display to existing workspace?