Signal Generators and Sources
to create and rate content, and to follow, bookmark, and share content with other members.
E5505A Down converter
Question asked by
on Dec 27, 2010
on Apr 14, 2011 by tabbott
Show 0 Likes
Please do let me know what is the main diffrences between E5505A and E5507A.
Advance thanks for your answer.
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Apr 14, 2011 11:50 AM
Hi Narasing -
The E5505A is a modular phase noise measurement solution. It consists of a PC controller with FFT digitizer and an N5500A test set. An optional component is an N5507A downconverter that converts microwave signals up to 26.5GHz down to a lower RF signal for comparisons with an RF reference source (there is no such model E5507A). There are several downconverters that work with the E5505A system, the N5507A, N5502A, and the N5501A, with the main difference being the upper frequency limit. For more information, see the E5505A Performance and Configuration Guide.
http://www.home.agilent.com/agilent/pro ... s.Overview
E5505A Performance and Configuration Guide:
http://www.home.agilent.com/agilent/pro ... ageMode=CG
Show 0 Likes
Retrieving data ...
Uncertainty with embedding fixtures
PNA-X B model Recall print colors command not working
Ports in Momentum: SMD vs TML
A Simple Method to Verify the Bandwidth of your Probe