to create and rate content, and to follow, bookmark, and share content with other members.
Question asked by
on Jan 2, 2013
on Jan 3, 2013 by Raclegg
Show 0 Likes
I am getting an Error 20, Line Down error on five of six N3300A electronic load mainframes that we have. Each of the N3300A mainframes has six N3302A modules and all are connected to 120VAC. What does this error mean?
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Jan 3, 2013 3:20 PM
First let me tell you what the failure means. I checked the code, it means that the mainframe is definitely having a problem communicating with the module(s). Possibly the module is dead or cabling is bad. If this is a new setup can you check that all the interconnect cables inside the mainframe that run between the modules are fully seated and properly lined up. Also make sure the pins are not bent inside the connectors in the modules. If you have 5 out of 6 mainframes it sounds more like a setup issue than a bad mainframe issue. If you are still having problems can you post a picture of the cabling with the top cover off of one of the bad units.
Show 0 Likes
Retrieving data ...
ask for a display ActiveX Control for ESA4402B
70871A 70872A Analyzer Measurement Personalities
noise figure measurement personality for 71209A
71209A Noise Figure personality