to create and rate content, and to follow, bookmark, and share content with other members.
How do I measure Chirpped Pulse's SSL and PCR
Question asked by
on Apr 2, 2007
on Apr 2, 2007 by bobmat
Show 0 Likes
Maxim Afanassiev, an Agilent Applications Engineer from Russia, wants to help a customer measure the quality of a linearly chirped pulse. The quality metrics are Side Lobe Level (SSL) and Pulse Compression Ratio (CPR). :?: [/b]
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Apr 2, 2007 10:45 AM
The 89601 Vector Signal Analyzer (VSA) software is a versatile tool that also featured User DSP Math which can be used for this measurement.
SSL is a quality metric and is the level of time side lobes in a compressed chirp after processing thru a matched filter. The matched filter can be made from an ideal version of the chirp made from Matlab. The VSA software's User Math then is used to cross-correlate the measured pulse against the ideal. The resulting compressed pulse is viewed in the logMag format and the time side lobes are measured.
The PCR is the ration of the uncompressed to compressed pulse width. The theoretical PCR is
PCR = pw * dev /k^2
pw is the uncompressed pulse width in sec
dev is the pp frequency deviation for a linear chirp
k is the window shape factor used in the measurement
k = 1.43 for a flat top window
See the enclosed Agilent Aerospace/Defense Symposium paper for the details of making this measurement.
Advanced Pulse Anallysis_Agilent AD Symposium 2004.pdf
Show 0 Likes
Retrieving data ...
Uncertainty with embedding fixtures
PNA-X B model Recall print colors command not working
Ports in Momentum: SMD vs TML
A Simple Method to Verify the Bandwidth of your Probe