to create and rate content, and to follow, bookmark, and share content with other members.
Webcast Q&A: Is internal attenuation a passive or active circuit?
Question asked by
on Mar 27, 2013
on Mar 27, 2013 by tabbott
Show 0 Likes
Questions and Answers to *Signal Analyzer Fundamentals and New Applications Webcast* on 13 March 2013
*Question:* Is internal attenuation a passive or active circuit?
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Mar 27, 2013 4:03 PM
*Answer:* We have two types of attenuators in our X-Series Analyzers. The mechanical attenuators are passive devices. The electronic attenuators are active devices. The primary attenuators in X-Series analyzers are mechanical, because mechanical attenuators are able to cover a wide frequency range, up to 26.5 or 50 GHz. For the low band path, X-Series also has an electronic attenuator. The advantage of the electronic attenuator is that it can switch faster than a mechanical attenuator.
Show 0 Likes
Retrieving data ...
Genesys USB Hardware Key driver
Mating Habits of Microwave and Millimeter Connectors
AEL syntax highlighting in Notepad++
IC-CAP PEL manual?
Default mesh settings