to create and rate content, and to follow, bookmark, and share content with other members.
Webcast Q&A: Tips on using 75 ohm input (CATV)
Question asked by
on Mar 27, 2013
on Mar 27, 2013 by tabbott
Show 0 Likes
Questions and Answers to *Signal Analyzer Fundamentals and New Applications Webcast* on 13 March 2013
*Question:* Tips on using 75 ohm input (CATV)
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Mar 27, 2013 3:33 PM
*Answer:* When making 75 ohm measurements, we recommend using a minimum loss pad to match the 75 ohm impedance to the 50 ohm impedance of the analyzer. A minimum loss pad has an insertion loss of 5.7 dB for which you can enter this loss into the analyzer for it to correct for. Also, you can choose under the Input – RF Input setting to use 75 ohm for the input Z correction. Please see the following FAQ: http://www.home.agilent.com/agilent/editorial.jspx?cc=US&lc=eng&ckey=529371&nid=-11143.0.00&id=529371
Show 0 Likes
Retrieving data ...
Genesys USB Hardware Key driver
Mating Habits of Microwave and Millimeter Connectors
AEL syntax highlighting in Notepad++
IC-CAP PEL manual?
Default mesh settings