to create and rate content, and to follow, bookmark, and share content with other members.
Webcast Q&A: How can I extend the dynamic range of my signal analyzer?
Question asked by
on Mar 27, 2013
on Mar 27, 2013 by tabbott
Show 0 Likes
Questions and Answers to *Signal Analyzer Fundamentals and New Applications Webcast* on 13 March 2013
*Question:* How can I extend the dynamic range of my signal analyzer?
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Mar 27, 2013 3:20 PM
*Answer:* The dynamic range of an analyzer is the ability to measure high level to low level signals. Second and Third Order products often limit the ability to measure low level signals in the presence of high level signals. To overcome this, it is necessary to increase input attenuation to reduce distortion products. As input attenuation is increased, the noise floor rises. To decrease the noise floor you can reduce the resolution bandwidth, which causes the sweep time to slow down by 1/RBW^2. For measuring low level signals, other ways to increase sensitivity include turning on the preamp and using noise floor extensions.
Show 0 Likes
Retrieving data ...
Issues connecting to M8190A
dsox3024 / msox3024 triggering
Transphorm Manages Layout Parasitic Effects Using ADS
Reading measurements without interrupting measurement
How to use Verilog A model in ADS