to create and rate content, and to follow, bookmark, and share content with other members.
HP 8592A frequency drift - Correcting sorry !!!
Question asked by
on Jun 7, 2012
on Jun 8, 2012 by tabbott
Show 0 Likes
When I put a stable signal of 100MHz in my HP 8592A with span of 10KHz the frequency drift in 3 KHz in both sides. Is a normal situation?
Edited by: euchuma on Jun 8, 2012 6:31 AM
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Jun 8, 2012 11:23 AM
The 8592A spectrum analyzer, which was manufactured in the 1980’s, has a free running 1st L.O., meaning it is not phase locked to any reference. What this means is the L.O. will drift and that makes the frequency accuracy of the instrument to not be very good. The 92A did have a feature called Signal Track that you may find useful. It is used to keep the signal at the center of the screen. Essentially, the signal track performs a continuous peak search, marker to center frequency, which will obviously change the center frequency of the analyzer as the 1st L.O. drifts. The frequency accuracy specification for the 92A was at least +/-5 MHz, plus other uncertainties like 2% of span, etc.. Typical instrument performance was better than this is most cases, but Agilent only warranted the published specification.
Show 0 Likes
Retrieving data ...
Issues connecting to M8190A
dsox3024 / msox3024 triggering
Transphorm Manages Layout Parasitic Effects Using ADS
Reading measurements without interrupting measurement
How to use Verilog A model in ADS