Data Acquisition and Switch Units
to create and rate content, and to follow, bookmark, and share content with other members.
34970 error 608
Question asked by
on Jun 24, 2011
on Jun 24, 2011 by lhornburg
Show 0 Likes
34970 mark in a self test error (608), swap the multimeter board to the other 34970 and pass the self test, how I can repair the multimeter board ? or should I check points?
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Jun 24, 2011 1:19 PM
It's hard to tell the root cause, it could just of been a poor connection but here is a more detailed description of what that error means:
Error 608 - Serial configuration readback failed This test re-sends the last
9 byte serial configuration data to all the serial path. The data is
then clocked back into A1U209 and compared against the original 9
bytes sent. A failure occurs if the data do not match.
There was an issue with some of the ribbon cables connected to the DMM board, see service note below, but I would expect other error messages and some fluctuating readings if that was the case.
Show 1 Like
Retrieving data ...
Thinking how to get money off Cash App without card and without net? Call customer care.
how to simulate the large siganl loop gain
Contact Facebook if you have problem getting a code for account activation?
Power calibration with use of receiver "A"
Importing EMPro circuit component definition from ADS