to create and rate content, and to follow, bookmark, and share content with other members.
record a spectral shift over time
Question asked by
on Mar 25, 2012
on Mar 26, 2012 by mkelly
Show 0 Likes
I want to record an interference peak shifting over time for my sensor's response time.
How can I record the response time using OSA Agilent 86140B?
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Mar 26, 2012 12:25 AM
It looks like you'll want to do this with a remote program to take the measurements at the desired time intervals and record the values.
For this you'll probably want to disable continuous triggering and then start the individual measurement sweeps with your program, as with these two commands:
OUTPUT @Osa;"init:cont off" ! turn repeat sweep mode off
OUTPUT @Osa;"init:imm" ! initiate a single sweep
The second command can be used for each measurement. Either the trace itself could then be stored or for example a peak marker could be used to locate the peak and then the value can be read out.
OUTPUT @Osa;"calc:mark1:max" ! set marker one to max
Another idea: If a faster update is needed than can be obtained with the repeated scans and readings, an alternative is to set the OSA to "zero span" by setting the span parameter to zero with the wavelength set to the flank of your peak. Then the trace will show the time dependence of the signal at this wavelength and will move up or down if the peak moves to lower or higher wavelength.
Show 0 Likes
Retrieving data ...
Doing Impedance matching by smith chart for S1P input file and see the resu
ael script and opening ADS for the first time
6000 series firmware - suggestions/wishlist....
Where is the DSO6000 firmware?