> From: Jaap Schuurmans <email@example.com>
> Subject: Re: Vrf: Reading HPGL from a talker into a file
> > > > > I would like to be able te read the HPGL produced by my HP8594E
> > > > > Spectrum Analyzer (or any other instrument with HPGL plotting
> > > > > capabilities), and store the results to a file. I would prefer a
> > > > > solution using Direct I/O.
Just triggered. It took a while.
I'm doing the same (at least I think it is the same) with a PM3382
combiscope from Philips. I didn't have much problem with it.
The direct I/O box looks like this:
- WRITE TEXT "HCOPY:DATA?" EOL ;query to scope
- READ BIN dummy BYTE ARRAY:2 ;two byte header thrown away
- READ TEXT output STR ;the actual output
- EXECUTE CLEAR
- EXECUTE LOCAL
while the Direct I/O configuration looks like:
READ TERMINATOR "
EOL SEQUENCE "
Is this of any help?
With kind regards,
Philips Electron Optics
Building AAE-room 21
5600 MD ACHT, the Netherlands
tel +31 40 2766947
fax +31 40 2766820