When we simulate our circuit in ADS software then error message is shown and error message is- 'MOSFET1'is an instance of an undefined model MOSFET1'. So tell me how I remove this error for simulation of our circuit. #simulation problem GShankaraVForumsadminandgenthAgilentModeratorDartster57tsilchia001stoyan_ganchev
- yingziwin likes alain001's question mixer broken for 8595E
- zizi1999 likes sdgrant's reply Re: S parameter sim of Wilkinson divider
- zizi1999 likes krishk24's question S parameter sim of Wilkinson divider
- zizi1999 likes mmm365's question circuit component pin number does not match EM layout pin number
- Filippos likes Svyatoslav's question Saving data of eye diagram/color grade scale using 86100 series
When I calibrate HP8753B (open, short, load, done), the text on the right side of the display turns upside down. Where can the problem be? I tried to replace the display module without success.
I have an issue with installing any program on the newly upgraded PNA-X N5242B with Windows 10 (upgrade from N5242A and Windows 7). For what ever reason when I click on any installation file I see spinning circle for couple second and nothing happen after that. No error , no warnings. I try to run the installs as administrator and still… Show more
Scope firmware for 90000 Series;Windows XP operating system It is running 1.20 now. upgrade to the latest software is 4.20
I need help the Scope software upgrade from XP version 1.20 to 4.20 It is having problems.failing last set to load scope application. Work Around the Known Issues
Does your scope have the Windows XP SP3 installed? That's now 'abandonware', and available on the web. You might also install the latest XP patches. Even though Microsoft stopped supporting XP quite a while ago, there are patches available if you make a registry change. Google 'Windows XP POSReady' for information. Even this is going away in…
In our lab, we have had sometimes strange behaving on 432A+478A-H76 mount. We measuring Vcomp and Vref voltages backside of the 432A using 3458A. Normaly when mount zeroing is fine etc, 3458A should show around 85mV when power meter DUT Ref-Out power is ON. Time after time even zeroing is perfect (less than +/- 200uV) and turn Ref-Out ON voltage… Show more
Hi, I have been searching a SCPI command for deleting the specific measurement on DSOS054A. In programming manual, I found how to add a measurement but I don't find any SCPI command to remove any measurement from the measurement table. There exist only :MEASure:CLEar or :MEASure:SCRatch commands which remove all the measurements added by… Show more
Hello, I currently have a 6675A DC power supply and have connected a constant resistance heater to it. The resistance of this heater is 12.65 ohm. On programming the output voltage to 6 V, the expected current draw from the supply should be around 0.5 A following Ohm's Law. However, with the output voltage set to 6 V, I first get a current of 3.5… Show more
Hi, I want to capture and parse the screen capture of the N6705B via SCPI commands to the controlling computer (computer running VBscript). How do I capture the screen and pass it/save it on the controlling computer? I know that the command "MMEM:EXP:DLOG "external:\data1.csv"" saves the screen capture to the USB drive attached to this power… Show more
how can the instrument automatically mark these three points （red circles）? For example (mark1,mark2,mark3). Then it's convenient for me to "CALC: MARK1: X?" read mark1's frequency and "CALC: mark1 :Y?" read mark1's power. "CALC: MARK2: X?" read mark2's frequency and "CALC: mark2 :Y?" read mark2's power. "CALC: MARK3: X?" read… Show more
Hi, this is my first question in Keysight community. I have met a problem in LTE demodulation: I have made a 20ms LTE 20MHz signal (sample frequency is 61.44MHz) in SystemVue. To verify the finite word length effect, I send the quantized data to VSA by building a SystemVue simulation system. When I demodulate the data in VSA, I can get the… Show more
Load more items
I have a E4438C,recently the unit started developing issues with the motherboad,i replace U901 AT45DB081B,it works however the serial# lose,and it display 617 configure error.What information does the flash contain?And how do i do?
Hi Lee, I'm afraid the software used to reprogram the flash memory is proprietary. The serial number needs to be loaded, files structures established, and an encrypted configuration file that contains model, hardware, and operating limits needs to be loaded. This can only be done at a Keysight Service Center.