Top & Trending
Register for the Community: If you don’t have a myKeysight account: Register for the Community by clicking here. If you already have a myKeysight account: Log in to your myKeysight account by clicking here. Login with your username and password. Click on “Add a Capability” in the upper right and select "Community".
I'm trying to measure precisely E4417A power meter's 50 MHz/1mW RF reference output using HP 432A power meter with 478A thermistor mount. I have followed 432A's user guide note how to make precision power measurement taking account those Vcomp and VRF voltages. I have one Agilent 34401A DVM for that purpose. Main problem is that those Vcomp and