Hello, I am trying to simulate MIMO_3DChannel model part. I want to get output at its output ports ht and HF. Can anyone please tell me how should I do it? Thanks in advance.
Hi, I would like to integrate automated measurements into my application. Are there commands available to read current and voltage from the handheld DMM U1273A? The Driver installed automatically on Windows 10 as a Prolific virtual com port. I was able to use the data logging software, but now I want to make a custom application. Any info… Show more
Hello all together I´m new in here and now I´ve got a question about a failure I have on a Hybrid Pin Card The failure is numbered 4512 Subtest 0 during the auto adjust " Control Card timeout waiting for TIC done " does anybody knew this failure?? what should I do to repair the card?? Regards Gaddy
Hi, may i know what the different between Series 3 and series 5 tester that using CET feature? i heard that need to add additional wiring?? do anyone have the information. thanks.
Hello, I have encountering issue on DGN report. The DGN report is just show simple report only, exp. Module 2 slot 11a FAILED. I want to see details report as below sample: Test 2348, Subtest 0, Module 3, Slot 1 FAILED. Module power supplies -PPM(-10 V) Low Limit : -1.05021500E+01 Result expected : -1.00000000E+01 High Limit : -9.49805100E+00… Show more
I was using N8485A sensor with N1913A for over 3 years. From last week the sensor calibration process encounters error "-231:data questionable". In fact when I connect the sensor to the internal 50 MHz 1mW source, the meter reads -31 dB instead of near 0dBm, but the reading value is totally stable. I also directly connected the sensor to the… Show more
Hello Does anyone have the sample Python script controls E4990A and gets trace data from it ? I am going to try USB and PyVISA to connect to E4990A. It would be helpful if someone shares the script with me. Thanks in advance. Toru