Edge-to-Edge measurements require Measure All Edges to be enabled. The documentation clearly states measurements will be made on all edges within the acquisition even those occurring off screen. Is the way to ensure that the Edge-to-Edge measurement is only made on what's on screen setting the memory depth to correspond to the displayed… Show more
When I connect my U2352A to Command Expert the connection process shows an IVI-C and IVI.net command set available. I have successfully connected using IVI-C and issued some commands. When I connect my U2352A to VeePro 9.33 (Windows 10) I can open the Advanced Instrument Properties box and go to the IVI-Com driver tab. However there the… Show more
Hi, i'm very beginner of ADS and would like to seek advice from any of experts here What i am trying to do is extracting XY coordinate from GDS file. There are a lot of circle pads and i want to export their XY coordinate to any human readable documentation such as *.txt file. The XY coordinate here stands for XY coordinate of center of a… Show more
本文作者：中山大学，陈彦伍 EMPro有先进的Python编程环境和丰富的内置函数库。本文通过范例介绍针对已有工程文件如何通过Python程式自动进行电磁仿真或参数扫描。使用该程式可以自动执行任意工程文件的仿真，避免一次又一次在EMPro UI里手动配置仿真所需的 配置项（如频率范围、求解器类型等）。 所用到的Python程式包含在范例工程文件中。
I saw "Demo Clock" as one of the options in the Calibration window's Aux (Output) pulldown menu. It's a 456.35Mhz signal, but I've never seen any reference to it on the Service Manual or the User Guide. Out of curiosity, is there any purpose for the "Demo Clock" signal? Is it used in the self-calibration process?
Hello, A 2017 Keysight video shows the use of Command Expert Sequences in VeePro. https://www.youtube.com/watch?v=lLr3IZor5B8 I have Command Expert installed on the same PC as VeePro (latest version). There is no "Command Expert Sequence" in VeePro I/O pull down menu. This contradicts the training video. How can I fix this? Regards,
I have an Agilent 8720ES that tests from 50MHz to 18GHz but my customer requires VSWR and Insertion Loss from DC to 18GHz. I looked for rental equipment to do the testing, but I'm getting quotes for equipment that has tests 1MHz to 20GHz. So how do I test DC to 18GHz on modern network analyzers?
Not quite DC but you can get to 30 kHz with FieldFox analyzers. Keysight FieldFox N991xA or B analyzers' Network Analyzer mode starts at 30 kHz. These units are generally available as rentals. Look for models like N9917A, N9918A, N9918B. Here is a list of Keysight's rental partners. How to Buy or Rent Electronic Measurement/United States |…