It would be nice if in a future firmware update <hint, hint> for the 2000-X Series that you could screen capture the snapshot measurements screen. One screen capture with all of the available info needed in one graphics file. Its just a hassle to capture screen measurements four settings at a time. Just an item for future consideration that I'm… Show more
Hello everybody, I have a problem with the #channel #simulation. I want to simulate a de-embeddet channel but I noticed that I get two different results although I simulate the same thing. I use the same touchstone file for the snp2 and d#e-embedding building block (ChannelSim1.jpg) and get the following result (Result1.jpg). However, if I… Show more
Opportunities for electrical and computer engineering (ECE) graduates seem to be virtually unlimited, with growth in health and medical devices, Internet of Things, consumer electronics, supercomputing, artificial intelligence, cybersecurity, smart energy, electric vehicles, robotics, cloud computing, augmented and virtual reality, 5G, machine… Show more
hi i try to communicate with KEYSIGHT 3458A i connected it to my computer with usb to usb A cable and than with prologix adaptor to gpib conect to the ketsight multimeter. i used with teraterm softwear to comunicate and send id? and "beep" commands but it is does not respond, what can i do?
Could you help me fix error code 740 on agilent data logger 34970a.
It’s official: Keysight Technologies is now certified as a Great Place to Work by the leading global authority on high trust, high-performance workplace cultures. Great Places to Work® surveyed our employees, who said that they enjoy a great atmosphere, great challenges, and great rewards. As a result, 92% are proud to tell others they work at… Show more
Hello Keysight! I got a brand new U1732C Handheld LCR Meter. It has a built in IR communication interface, for which I also have the appropriate cable. When setting the cable to 9600 baud, I successfully could send "*IDN?" and got the reply "Keysight Technologies,U1732C,MY12345678,03.06". However, even an extensive search did not reveal… Show more
Hi Johann, based on answer from HansDuo ( 2011-11-12 12:16) and some tests that I have done, this is result below. Indentification of LCR meter -> *IDN? <- Keysight Technologies,U1733C,MY58020034,03.06 Set Frequency: 100, 120, 1k, 10k, 100k -> FREQ 100 (example: set frequency of LCR meter at 100Hz) -> FREQ 10k (example: set frequency of…