can we log DC current surge in micro ampere for initial 20ms using @34972A? If yes please explain how to do it.
the 34972A with the 34901A module, can read DC current in μA on the 10mA range with a maximum speed of 500 readings/sec or 2ms/sample (scan one channel only), however the measurements will be more noisy because the device will be in 4½ digit 15 bits. To answer correctly, give more details: you say "initial 20ms", is it the period of measurement or the width of the surge?
Retrieving data ...