Top & Trending
Hi! I just got this great (if working) bench multimeter. It is faulty and there's a lot of different fault codes. 602,603,604,606,611,612,613,614,615,616 etc.. what is wrong with this meter?? It's old agilent one so there's no warranty. Is it simple thing to fix because lot of errors and one thing that can cause all of this errors.. My
Is there a DMM that can replace the 3458A that iws code compatible or has an emulation mode for that?
I have a system with three 3458A meters in it. But in the change from Windows XP to Windows 7, I have had problems with these meters being very fragile in terms of GPIB communication. I don't think we need the accuracy of the 3458A so I'm wondering if I change to another DMM (like 34470A) will I have to rewrite the library for the DMMs. From