Top & Trending
I am trying to learn writing automated test setup Matlab codes for keysight instruments. I wrote some simple codes for different instruments and they run perfectly. As a next step, I am trying to generate a signal from Agilent E4422b signal generator and reading it in Matlab from the Agilent E4405B spectrum analyzer. After
Is PXA's (9030A) averaging speed in FFT mode dependent on signal power level? I am seeing slower average measurement when power is closer to the Reference level but it gets much faster once the power level decreases and goes further below Ref level. Thanks. Steve Edit: An E-band external smart mixer was used if that makes any difference.