Top & Trending
Hello, I recently acquired an old 8561B w/ prefix 3017A. Powers on and completes the self tests but with errors 301 317 and 334 (suspect a dead YTO which is a common and costly issue or hopefully a dead LODA). Currently I've been going over the 8561/3 E/EC manual as a general reference. I was wondering if anyone here or out there
Hi, I bought a used DSOX2002A. I updated it to firmware 2.35 and I started the 30 days of evaluation. The scope functions are OK. On the demo PIN all the signals are OK but I can't use the waveform generator function. When I switch on the scope I have the message "Too high a voltage is being applied to the waveform generator BNC".