Top & Trending
I have a new-to-me DSA90604A with the latest firmware, and it passes all self-tests and self-calibrations. I've noticed that when it is displaying a steady waveform (such as its own calibrator output or a 10 MHz GPSDO), it seems to "stutter." That is, the waveform will freeze for a brief fraction of a second, then go back to continuously updating.
Dear all, I'm facing the current strange problem. I have a Keysight 33120A AWG connected via a Keysight GPIB-to-USB interface of type 82357B to a 64 bit laptop running on Microsoft Windows Enterprise SP1. I have on that laptop several accounts for different users. I have installed the Keysight Instrument Control Bundle and the 33120A