How to transfer file to E5052B instrument from PC connected via USB using SCPI commands? I have tried :FORMat:DATA ASCii :FORMat:BORDer NORMal :MMEMory:DATA "F:\a.txt" "SAMPLE" Instrument gives error "invalid seperator"
Hi, I have at work one 8590B that has suffer of boring problem of backup battery dead. I have replaced the battery (both lithium and supercap) and I tryed to recalibrate the instrument. When I performed early attempts, the 8590 says that not found the calibration signal. In effect it was displayed with a considerable offset. I have tryed a…(Show moreShow less)
Hi, If you cannot see a synchronized image, you will have to go to the blind menus based on USER 08590-90001 on page 212. and Verification manual 5958-7098 page 413 for pass code. The 6 menu choices correspond to the 6 soft menu buttons. You must therefore go to the DEFAULT CAL DATA menu. to access at this menu you need a code Regards…
Hi I have HP8594A spectrum analyzer and i am trying to back up Correction Contansts but analyzer appears to reset when Edit Fflatness is initiated. More info : Following service manuals instructions, using -37 Hz password, "srvc" appears on the left down corner, next if i push "VERIFY TIMEBASE" the Timebase Number appears and everything seems to…(Show moreShow less)
Hi engineer, My 8596E spectrum analyzer suddenly failed. The fault is manifested as the interference of various random frequencies on the frequency spectrum, similar to the interference of a single carrier. In order to troubleshoot, I replaced the intact A8 Power Supply, A7 Analog Interface, A15 Motherboard, A9, A11, A12, A13, A14, A25 and all…(Show moreShow less)
We have N8975A that we would like to measure noise figure for LNA amplifier at 36 GHz. As we know, we can use Option K40 (downconverter + LO) for this. My question is: Is there an alternative to Option K40 to make noise figure measurement for LNA Amplifier at 36GHz? I believe an Dedicated LO and a Mixer should work. Please suggest…(Show moreShow less)
We would like to measure a LNA with frequency range 25GHz to 28GHz and would like to use Downconverter Mode on our N8973A noise figure analyzer. Could you recommend a Downconverter for this application? Does Keysight sell this Downconverter for this application? Does Block downconverters that are used with N8975A will work with N8973A? …(Show moreShow less)