Does anyone know appropriate reference for the legacy definition of RINOMA? There is much paper explaining what IEEE RINOMA is. But I could not find the paper describing the legacy equation of RINOMA used in FlexDCA.
On my MSO9404A ( ver. 06.40.01001), how do I set a fixed number of captures so that when I start the scope it will take (say 100) captures and stop or pause keeping the last (or infinite persistence) waveform as well as the measurements and statistics?
The Segmented sampling mode (see Setup > Acquisition...) with infinite persistence set and measurements added will do this. I also thought the macro recorder (of a Single run), copied 100 times, and playing back might work, but if the speed is Fast, some Single runs get skipped. A remote program that executes 100 :DIGitize SCPI commands will also…
This is a new scope for my bench. I currently have a Rigol scope using Ultrascope software to display screens etc to my computer. I would like to do something similar using this oscilloscope. Ideally, I would like to use the same software to handle both oscilloscopes. Is there a low cost, or freeware application that can be used to accomplish…(Show moreShow less)
I have, Keysight InfiniiVision 2000 X-Series Oscilloscope, model DSO-X 2024A AGILENT Technologies, I am using Digitize, then waveform preamble to capture the data, using ASCII format (because the precision is really important) , I am using Visa for communication and programming in QT C++. Now the question is that when I use byte I lose…(Show moreShow less)
Is there a way to convert a learn string for a 6000 series scoop to a 3000 series scoop ? Or convert any learn string to a set of SCPI commands ? We have a lot of learn strings from a 6000 series scoop and we need to "upgrade" them to a similar 3000 series scoop.
Hi, I recently try to do jitter measurement on DCA-X module 86108 for a 53G PAM4 signal. I have a deembedding function to remove a S4P file, and followed by a CTLE transfer function that applies 4dB peaking. In order to use those, I had pattern locked to PRBS13. Under Eye mode, I see that the eye is pretty open with 60+ mV eye height and…(Show moreShow less)
Hi, I'm using the DSO9404A oscilloscope as a pulse counter. I trying to build statistics from measuring the mean of many measurements of pulse counting. My problem is that the scope only takes under account measurements when there is at least 1 pulse inside the measured interval. When it sees a measurement without any pulse in it, it adds a…(Show moreShow less)
Hi, I have DSOS054A oscilloscope. In the datasheet, the dynamic range (SnR) is give 108 dB. But as I know, the dynamic range formula is SNR=6.02N+1.76dB where N is ADC bit number. DSOS054 has 10 bit ADC. As I understand from "Keysight Technologies Evaluating Oscilloscope Signal Integrity" application note, it increases to 12 bit with using the…(Show moreShow less)