Log in to follow, share, and participate in this community. Hello I had to drill a hole in the wall in potential line of the power line. So I thought of the VSense function of my Keysight - U1282A to check if there are wires hidden in the wall where I want to drill. ... Hello I have a couple of U1272A devices, that I want to communitcate with using C code. I can see them on my device manager as a com ports. My question is, where I can find a list of commands to get re... Hi at all, does anyone have detailed information about the Selftest of the L4534A Digitizer? How does the self test works? The overvoltage protection test is sometmes failed, if the unit is in warm con... I there. I have a 3458A multimeter that we have repaired, after some errors showing on the display caused by the Dallas NVRAMs and Multislope Rundown and ADC Errors and so on. We switched the 2 dalas NVrams and the on... Hello, I am attempting to create a program that will setup measurement of the 34465A and continuously send data to PC without interrupting the measurement. I set up measurement with an infinite trigger, th... Hello. I am measuring voltage drop on several DUTs at externally triggered intervals with my Keysight 34980a DAQ and 34922a Multiplexer Module. I want to add operating current measurements, but the Multiplexer Module ... Hello, I need to measure the ambient temperature with the 34420A and I was planning to use a thermistor, but I'm not sure what Beta value the thermistor should have. Please advise. T... Hi I have purchased a 34465A DMM with the transportable license, Idk how to activate this license on my computer. Another question is what is the certificated number when I am creating my account? I only... I've bought a 4339B, which I have not got yet. There are several different connectors on the front, and I could not determine what they are The interlock connector - perhaps a Lemo of some type ??? The grou... Hi, I'm trying to take readings quickly via my c# app, but testing out in Command Expert first, my last command causes a SCPI Error "211 Trigger Ignored", I can't find any info about why it was ignored, has anyone res... Hi, I'm building an app using the IVI-COM drivers using the direct driver instantiation. I am successfully connecting to two DVM's and talking single readings from both, however I now want to trigger with enco... Hi all, I am a Python programmer but I am completely new to this product and to SCPI. I can successfully connect and read data, change display etc but I am having trouble configuring / seeing my s... Hi, I have been using the 34461A to measure voltage from an optical power meter. But I now need to measure outputs of two such power meters. Is it possible to use the front and back panel I/Os at the same... On page 3 of the Digital Multimeters 34460A, 34461A, 34465A (6½ digit), 34470A (7½ digit) Data Sheet it indicates that the maximum reading rate for DC voltage, current and resistance for the 34465A with... Hello, I wonder if anybody else has run into this problem, or if an explanation could be found. When using the 34465A for sampling an input signal (using the DIGITIZE and MEM options), the sample rate is consist... Hi, I am measuring the current consumption of TI CC2640R2F Launchpad which is powered by 2 AA Batteries. As is it works as expected. However when I connect the Keysight U1232A to measure current, the TI CC2640... So, I've built my app which uses the 34410a dvm drivers, works perfectly from my dev machine, but on another machine, it doesn't run or throw error on event viewer. I guess it's missing something, but the release buil... hi i try to communicate with KEYSIGHT 3458A i connected it to my computer with usb to usb A cable and than with prologix adaptor to gpib conect to the ketsight multimeter. i used with teraterm softwear to comunicate a... Hello Keysight! I got a brand new U1732C Handheld LCR Meter. It has a built in IR communication interface, for which I also have the appropriate cable. When setting the cable to 9600 baud, I successfully could... I tried to call support but I am in Mexico and the tel. number that used to work and is on your webpage does not work anymore. Is there any voice support directly to Keysight from Mexico?