Log in to follow, share, and participate in this community. Have a U1241B handheld multimeter. Using a function generator with 50ohm output impedance to output a 1Vpp 1kHz sinewave. Then I have a simple voltage divider of 100ohm and 10kohm connected to the func gen out... Hi there, I could not change default setup on my U1241C, any answer for that? Thanks. I am not able to start a measurement with my U1242B multimeter and the most recent data logging software (Keysight Handheld Multimeter Logger v22.214.171.124, version 3.1.51130.01. Firmware version of my U1242B:, MY55100300... Hello, I'm interested in an insulation resistance tester U1451A. But it is not clear for me, what are the measurement ranges. I assumed that the ranges are 6MΩ 60MΩ 600MΩ 6GΩ 60GΩ but this ... I have aU1241B multimeter which is under a year old and from new I have been wondering about the temperature reading. When no sensor is connected the reading is 5 degrees C off. The room temperature is about 23C and t... Hi all, I have been using this clamp meter for around 2 years. At most of the time i use it to measure DC and AC current on my product. When I'm measuring current of a DC circuitry, I will press Shift to ... I own an older U1273A Multimeter (Agilent Branded) running the original firmware. I have an existing U1173A USB to IR adapter that used to work on a Windows XP or 7 machine. I now have a Windows 10 machine. I am able ... When measuring a resistance on the meter, I get the following: resistance mode 1.06ohms (1 ohm resistor, 1% ) seems reasonable, leads were not nulled If I hit dual display I get the following mV ... Setup is from a HP33120A arb generator, 2 V p-p, 60 Hz, +0.500 VDC offset. Meter set to auto mode with normal 5-1/2 digit display when powered up. The display does not auto-switch to read the 0.500 volts. ... In the 34461A DMM in the table "performance versus measurement speed", we can read for a 0.2 PLC 100 Readings/s, but where does it come from? Because 3 ms of integration time + 1ms of delay (immediate sample) gi... I have multiple 34401A that we use for automated testing and recently noticed that the value is a short when nothing is connected(including leads) and the ranged is fixed at 100 ohms. This includes front an... I just received my meter and like it a lot. I don't understand how to set up the bar meter as a null meter. if I set it up as -5 to 0 to +5 the bar is from the left. So the bar grows from the -5 to 0 when the value ... I am testing some DC2DC modules from a PSU. Is it possible to connect PSU to PCBs through 34461A (current measurement mode) and limit the current? I couldn't see any option to limit the current. Hi, I have a problem with my 34410A: everything is OK except Ohmmeter.I tested with 0.01% resistor, 100, 1k, 10k, 100k, 1Meg and 10Meg and used my 34970A for DC current measurement (equivalent to 34401A). If I ... I'ver been trying for the past several days to download the latest firmware for my U1253B DMM (version 2.26), but every time I click on the download link I get the message: "Access Restriction. An Unexpected Error Occ... I use the 34401A DMM and connect it via GPIB to my computer. Two questions: What is the maximum cable length between the DMM and my computer with this type of connection? How close can a GPIB cable run next to a 230... Hi, I'm doing continuous (every 0.5s) current measurements with a 34461A on an Android device. The multimeter somehow interferes with the device so it reboots. For one device it's when it's booting, for another ... I have a new project where I need to grab a voltage from a 34410A DMM (via USB?, other possibilities better?) and a number string from a USB micro-controller which appears as a "virtual com port via ... I started adjusting this 3458A and looked at the temperatures: ... Is there a way to adjust the display brightness on the 3458A multimeter. Sorry, this thread looked closest so here I am.