Log in to follow, share, and participate in this community. This is an observation about the behaviour of the 34465A DMM which is different than the 34411A predecessor. Both DMMs have the DIGITIZING option (standard with the 34411A) and 2M Memory options fitted. When the 344... Hello I had to drill a hole in the wall in potential line of the power line. So I thought of the VSense function of my Keysight - U1282A to check if there are wires hidden in the wall where I want to drill. ... Hello - I have an Agilent 3458A unit (Malaysian built) and want to extend memory with option -001. Normally one needs to replace PAL device U102 with a modified logic content ... On the Keysight website - the... **** Brand digital 31/2 digits Insulation Tester model 954, 5kv multi-range (1kv-2.5kv-5kv) and Insulation up to 200 G???, having a lot of features like Ac Voltage measurement up to 600V, Phase sequence test up to 100... Could you please tell me the power adapter specifications of U1252B multimeter? Is it just a regular 24 VDC power adapter or is it a special charger? Hi, I am using 34401A for 4-wire resistance measurement at the point of contact between gold-plated copper contacts. These contacts are just physically touching under a normal force which varies between 1 to 100... Hello. I am measuring voltage drop on several DUTs at externally triggered intervals with my Keysight 34980a DAQ and 34922a Multiplexer Module. I want to add operating current measurements, but the Multiplexer Module ... Hello, I need to measure the ambient temperature with the 34420A and I was planning to use a thermistor, but I'm not sure what Beta value the thermistor should have. Please advise. T... I've bought a 4339B, which I have not got yet. There are several different connectors on the front, and I could not determine what they are The interlock connector - perhaps a Lemo of some type ??? The grou... Hi, I have been using the 34461A to measure voltage from an optical power meter. But I now need to measure outputs of two such power meters. Is it possible to use the front and back panel I/Os at the same... On page 3 of the Digital Multimeters 34460A, 34461A, 34465A (6½ digit), 34470A (7½ digit) Data Sheet it indicates that the maximum reading rate for DC voltage, current and resistance for the 34465A with... Hello, I wonder if anybody else has run into this problem, or if an explanation could be found. When using the 34465A for sampling an input signal (using the DIGITIZE and MEM options), the sample rate is consist... I tried to call support but I am in Mexico and the tel. number that used to work and is on your webpage does not work anymore. Is there any voice support directly to Keysight from Mexico? Hi, I've had a look at this post, (Win10 support for logging s/w ), for additional guidance but still can't get my software to run properly, this is on both an up to date Windows 10 and a Windows 8.1 install. Ba... Hello and a happy new year, I have a U1253B multimeter for 5 years now that I love to use because of the oled display. I always had the brightness to the lowest setting. Suddenly the display ... Any typ of 3D drawing of the 34450a or look alike. I have a U1241C DMM. My problem is with it only measuring AC current in A position. Theoretically it allow 10Amp range with a 0.001 resolution. When I switch to this mode to measure a little current, like a 230V/25W b... This video shows you how to setup the 3458A DMM to use the E2308A Thermistor probe directly so it converts resistance to temperature through the use of built in calculations. Video: http://www.youtube.com/watch?v=LlN... Have a U1241B handheld multimeter. Using a function generator with 50ohm output impedance to output a 1Vpp 1kHz sinewave. Then I have a simple voltage divider of 100ohm and 10kohm connected to the func gen out... Hi there, I could not change default setup on my U1241C, any answer for that? Thanks.